Chapters Table Of Contents Type the SCPI commands at the prompt. Rack Installation Do not block the air intake at the front, or the exhaust at the rear of the unit. Make sure that the connector plug is securely inserted into the connector body. For a description of these parameters, refer to the previous section. Measurement Techniques Verification and Calibration Appendix B Measurement Techniques Electronic Load Many of the test procedures require the use of a variable load capable of dissipating the required power.
|Date Added:||17 January 2005|
|File Size:||37.52 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
If the power supply fails any of the tests or if abnormal test results are obtained, try calibrating the unit.
Sales and Service Office for service and Configure the following LAN address parameters. The enabled state is On 1 ; the disabled state is Off 0. The voltage is measured before and after the trigger. All compatibility commands are accepted; however, some commands do nothing.
Agilent Technologies N5700 Power Supply User Manual
Appendix C Service Error Messages Make sure that the connector plug is securely inserted into the connector body. If you are using an Agilent A, you can set up the voltmeter to do this automatically. Press n5700 hold the LAN button for three seconds. The J5700 address is: Operating the Power Supply Remotely 4 previous command in the message up to and including the last colon separator. Connect each load to the distribution terminals separately. Device Clear also prepares the power supply to accept a new command string.
AGILENT TECHNOLOGIES N SERIES USER MANUAL Pdf Download.
Language Reference 5 Source Commands Source commands program the voltage, current, triggered, and protection functions. These are located under the Settings tab.
Measurement Techniques Aggilent and Calibration Appendix B Measurement Techniques Electronic Load Many of the test procedures require the use of a variable load capable of dissipating the required power.
Language Reference 5 System Commands System commands control system functions that are not directly related to output control, measurement, or status functions.
Keysight Technologies / Agilent Technologies N5700
AC peaks of the output current ripple, several DC measurements should be made and averaged. SCPI commands are based on a hierarchical structure, also known as a agilsnt system. Page Index verification equipment Rack Installation Do not block the air intake at the front, n500 the exhaust at the rear of the unit.
Operating the Power Supply Locally 3 To re-enable the output after it has shut down, you must disable the Shut-Off signal.
Local Sensing Installation 2 The J2 connector plug specifications are as follows: Page 98 Let the oscilloscope run for a few seconds to generate enough measurement points. In Safe-Start mode the Shut-Off function is latched.
Verification These procedures calibrate the power supply. These procedures calibrate the power supply.
Agilent Technologies System DC N User Manual | pages
Page 43 Operating the Power Supply Locally 3 The J1 connector also provides monitoring signals for the output voltage and output current. Place the locknut inside the AC input cover with the flat side of the nut against the cover. Connect to appropriate voltage source. Appendix D Compatibility Differences — In General The following table documents the general differences between the way Compatibility commands work on the Agilent N power supplies and the way they worked on the Agilent xA power supplies.
For a description of these parameters, refer to the previous section. All settings NOTE commands have a corresponding query. On the Agilent Infiniium scope, the maximum peak-to-peak voltage measurement is indicated at the bottom of the screen on the right side.